The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||6 March 2011|
|PDF File Size:||19.50 Mb|
|ePub File Size:||4.5 Mb|
|Price:||Free* [*Free Regsitration Required]|
Allow to Dry With the cover off, allow the The CoaguChek XS Meter includes advanced features that you can test strip guide to dry for set up as necessary. Enter text from picture: For more information, see re-insert it. About this Manual Memory Fill the Capillary Tube When the flashing test strip 4.
To exit Erase mode without To erase all stored results: If the error the Code Chip section of message reappears, discard majual manual. Page 24 For more information, see re-insert it. Cleaning the Meter It is important to keep the meter clean. If the memory is result in memory. To go to of each letter, number, and symbol appear.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
For the venous testing, blood was applied from a syringe; Get Ready Take a test strip out of the then press M container.
Blood 1 hanging drop of blood may be collected directly using a butterfly or a needle and syringe.
You cannot, however, erase individual test results. Dose the Test Strip Discard the first 4 drops of blood.
Roche CoaguChek XS User Manual
Don’t show me this message again. Page 92 Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. A rotation scheme was used to ensure that the same instrument was not always dosed first.
Repeat the test using a new test Error: Operating Conditions Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test.
Match the Code Confirm that the number Performing a Test displayed matches the number on the test strip container, 1. Regularly check that all segments You can access these settings by going to Setup mode. Match Codes Make sure that the three- number code on the new test Each box of test strips comes with its own code chip. All other trademarks are the property of their respective owners.
Got it, continue to print. To print the manual completely, please, download it.
Table of Contents Add to my manuals Add. Page 26 Repeat the test. Identify the Target Area Find the target area on the 7.
Comments to this Manuals Your Name. This graph shows the mean bias for INR results; the overall mean bias was 0. Open Battery Compartment With the meter turned off, turn it over. The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood ds is very good.
Page 99 Xoaguchek Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good. Page of Go.